Portable Optical Power Meter(-70~+6dBm, 850/980/1300/1310/1490/1550/1625nm, with FC/SC Connector)
The OPM-101 is a new high-precision portable optical power meter. It is a compact and an easy-to-use testing instrument for optical fiber networks, which can be used for absolute optical power measurements as well as for relative loss measurements in optical fibers. It is compatible with SC/FC/ST with the most popular interchangeable interface on the market.
|Product Series||Optical Power Meter||Optcore Part Number||OPM-101|
|Measurement Range||-70 ~ +6dBm||Resolution(dB)||0.01|
|Test Accuracy||±0.2dBm||Calibrated Wavelength||850/980/1300/1310/1490/1550/1625nm|
|Connector||2.5mm FC/SC Connector||Dimension||190x90x45mm|
|Battery type||Alkaline Battery(3 x AA 1.5V batteries)||Battery Life||≥ 50 hours|
|Environment||Operating: -10°C to 60°C|
Storage: -20°C to 70°C
1 x User Manual
Select the probe in the wavelength range that matches the source to be tested. If the light to be measured is output by the movable connector, clean the end face of the movable connector. If it is a bare fiber, make a flat end face perpendicular to the axis. Vertical pair The quasi-sensor mirror, or the corresponding fiber optic adapter and connector adder, can be directly coupled to the probe with the active connector after the fiber is inserted into the adapter and the end face is processed.
How to Maintain
Always keep the sensor end face clean. Once the optical power meter is not in use, immediately put a dust cap on it to protect the end face from cleaning, prevent the long-term exposure to dust in the air and cause measurement errors. Carefully insert and remove the optical adapter connector to avoid scratches on the port. If you do not need it for a long time, please remove the battery to prevent the battery from getting wet and affect its measurement.
Datasheet / Specs